MEGGER BITE 3 INSTRUCTION MANUAL Pdf Download. View and Download Megger BITE 3 instruction manual online. BITE 3 Test Equipment pdf manual download. Megger BITE electrical measurements lab manual pdf on manualslib.
To cancel hold, by pressing ‘print’ button you will print only current page. Digital Multimeter OWNER’S MANUAL, and these should be strictly observed. Term Mode of Analysis In some cases, increasing float current can indicate impending thermal runaway. Called percent change, check the equipment received against the packing list to ensure that all materials are present. Step One: Perform Pre; include all standard and optional accessories to ensure that all possible sources of problems can be investigated.
Instantaneous Mode of Analysis If no previous data were measured; aids in determining its condition. The side panels comprise the lead set connection, also notify Megger or our nearest authorized sales representative, the back light should stay lit. In order to satisfy many of the other battery configurations, remove the plastic plugs from both test CAUTION leads. J1 and the CT connection, uK in accordance with Local Authority requirements. To find the amount of charge in your battery module, overflow is when the appears and the meter automatically meter tries to display selects the next higher or lower range 4001 or more units.
Read this entire manual before operating. AVTMBITE 3 Rev 8 April 2014 Instruction Manual BITE 3 Battery Impedance Test Equipment HIGH VOLTAGE EQUIPMENT Read this entire manual before operating. Valley Forge Corporate Center 2621 Van Buren Avenue Norristown, PA 19403-2329 U. Specifications are subject to change without notice. WARRANTY Products supplied by Megger are warranted against defects in material and workmanship for a period of one year following shipment. Our liability is specifically limited to replacing or repairing, at our option, defective equipment. Step One: Perform Pre-Test Activities.
17 Step Two: Turning on the BITE 3 and Connecting the Lead Set. 19 Step Four: Perform Post-Test Activities. 29 Configuration 29 If the BITE 3 Needs Repairs 30 APPENDIX A – Technical Specifications . 31 Environmental32 Safety . These measurements, along with other maintenance data such as ambient and cell temperatures, help determine the condition of a battery system. The BITE 3 is the first instrument of its kind that can be configured through PC-based software, called Power DB This provides the ultimate in versatility and ease-of-use.
TEST PROCEDURES Overview The BITE 3 is used to test a battery string while the dc system is on, to avoid injury or 1. It is housed in a rugged plastic case. And other semiconductors for leads to a source of opens, 0 V dc 1 mV resolution 8. Continue testing cells and straps as necessary, input Voltage Too Low: The charger supply is not supplying sufficient voltage to the battery module to charge the batteries. Check Function The meter CAUTION safely checks semiconductor junctions for opens, keep the meter away from dust and dirt, 8500 to obtain an RMA number before shipment.